Glove Box Each Day Upkeep Guideline
one) Fill in the use history in time immediately after running the glove box, including "user, h2o oxygen price, what to do"
2) When replacing gas cylinders in the glove box, the pressure is established to the default price ± 3mbar
three) The every day performing strain is set according to the working comfort. When not in use for a lengthy time, the stress is set to 1-6mbar
four) Consistently test the don of the gloves, look at the oil level of the vacuum pump (involving the upstream and downstream markings), and increase the pump oil appropriately
5) Thoroughly clean the water probe the moment each individual 2 to 3 months to be certain accurate detection. (Also have to have to refer to the true situation of the atmosphere in unique user containers
six) If you frequently use natural and organic solvents in the glove box, it is encouraged that you on a regular basis replace the natural solvent adsorbent product, not afterwards than one calendar year at the latest
seven) Swap the filter ingredient frequently, no afterwards than two yrs
eight) In accordance to the true use of your glove box, decide to switch the vacuum pump oil cycle, this kind of as black and yellow pump oil, water or extra organic and natural solvents in the pump oil. The oil alternative cycle should really not exceed 2 yrs at the newest
nine) If a massive quantity of organic solvent is inevitably uncovered throughout the experiment, the cycle desires to be stopped for the experiment. Immediately after the experiment, 1st thoroughly clean and then start the cycle
ten) When the surroundings in the box is harmful or destructive gasoline, the exhaust port of the vacuum pump, the exhaust port of the regeneration gas and innovative technology glovebox the exhaust port of the cleaning valve need to be exhausted to the out of doors with an exhaust pipe (when exhaust is not authorized to be outdoor, the exhaust gas treatment ought to be completed initial).